The camera reverts to the sleeping mode when it is the charge station. Setting Contrast Adjusts the contrast value from 1 to 7. Setting 2-level capture Works as a 2-level button by default. Select the “Activate Camera” button within the acquisition module. Comment button Displays the comment and tooth number functions.
|Date Added:||9 June 2013|
|File Size:||29.6 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
System Components The KODAK intraoral camera consists of a camera that captures video or still images when connected wirelessly or wired to a computer or video monitor.
KODAK USER MANUAL Pdf Download.
Brightness Adjusts the brightness value from 1 to 7. Such action may cause an electric shock or a malfunction of the unit. Hygiene And Disinfection Use a clean, lint-free cloth to thoroughly wipe the camera with a disinfecting solution, being careful not to intraoarl too much liquid.
The Acquisition panel switches to the Review window, and the captured image is displayed in the Preview screen. Camera Indicator Battery Indicator To use the charge station as a stand-alone device, follow these steps: Go to the computer connected to the active docking station where work will be performed.
Used Dental Intraoral Cameras | Kodak 1500
Insert the camera or battery into the charge station. Hold the sheath in place on the camera by gently pressing the white tab against the camera with your thumb or finger.
Connect the camera USB cable to the docking station 1. When image acquisition is completed, remove and dispose of the protective sheath.
Figure 19 Active Camera Note: Wired Camera Press all the way down to capture an image. The camera has been designed to carry out the following functions: Remove the top cover from iodak docking station. The power indicator on the front of the docking station turns green. Using the switch at the bottom of the docking station, switch the output to the correct video output: WiFi indicator Solid blue indicates the connection is working properly; blinking blue indicates a communications problem.
Press all the way down to capture an image.
CS from Carestream Dental | Dentalcompare: Top Products. Best Practices.
Page 23 Displays the next or previous images in the Review window. Press the Shutter Release button halfway down and release it to return to the Capture window and live video.
Carestream Health is a trademark of Carestream Health, Inc. You can use the charge station as a stand-alone unit or attach it to the docking station. Select Install the software automatically Recommended and click Next.
Don’t show me this message again.
Carestream Kodak 1500 Dental Intraoral Camera Twain DEXcam Gendex Sopro
Connecting the Antenna to the Docking Station To connect connect the antenna to the docking station, screw the antenna into the connection on the right side of the docking station. Got it, continue to print. Connect the charge station power cable to the power input. The Found New Hardware Wizard window opens. Page 38 Figure 19 Active Camera Note: Using the Camera with a Protective Sheath The protective sheath covers the camera and provides a sanitary shield for the patient.
Without the SD card, or if the card is full, the docking station can store the last 12 images. The following figures illustrate the idle and active camera status.
Comments to this Manuals Your Name. Page 30 Insert the SD card into the socket on the front of the docking station.